To ensure the normal operation of the Somatos-Mini analyzer and obtain accurate measurement results, it is necessary to thoroughly wash the flask and capillary of the analyzer every day after the end of work. For washing, we strongly recommend using the Reaktiv-1 detergent specially developed by Sibagropribor, which effectively removes protein and milk stone residues.
for the use of reagent No. 1
1. Rinse the flask and capillary of the analyzer with water heated to 40 °C.
2. Heat water to 50 °C, dilute reagent No. 1 in it (1 g per 100 ml of water).
Rinse the analyzer flask and capillary by changing the washing liquid three times.
3. Rinse the analyzer flask and capillary with clean water until the water is clear.